Technical Information > Slip Resistance Test Methods
AS/NZS 4586 Slip resistance classification of pedestrian surface materials outlines four test methods to classify and assess the frictional properties of floor surfaces and include the following slip resistance test methods:
- Wet pendulum slip resistance test
- Dry floor friction slip resistance test
- Wet barefoot slip resistance test
- Oil wet ramp slip resistance test
WET PENDULUM SLIP RESISTANCE TEST
The wet pendulum slip resistance test (AS/NZS 4586 Appendix A) is generally conducted using a Wessex or Munro - Stanley London Pendulum Friction Tester. This pendulum device is portable and consists of a weighted foot with a test slider that swings down and slides across the surface wetted with water. The weighted foot comprises a spring loaded rubber test slider that exerts a prescribed force over the specimen as it slides across the surface. The results of expressed as a British Pendulum Number (BPN) and classified according to AS/NZS 4586 as shown in the table below.
AS/NZS 4586 classifies the surface based on the mean BPN where the Australian Standard for onsite slip testing AS/NZS 4663 expresses the result in terms of BPN. AS/NZS 4663 also provides a notional contribution of the floor surface to the risk of slipping when water wet as given below.
While there are 2 rubber slider materials that may be used, Four S (simulated standard shoe sole) is generally accepted as the material to assess the slip resistance for the general public wearing suitable footwear. TRL (Transport Research Laboratory) rubber is predominantly used for highly slip resistant and profiled surfaces. Research has indicated that TRL rubber may provide a better indication of barefoot slip resistance due to the softer yielding nature of the material, being more comparable to human skin than Four S rubber.
|Pendulum Mean BPN|
AS/NZS 4586 Classification
AS/NZS 4663 Notional* contribution of the floor surface to the risk of slipping when wet
|> 54||> 44||V||Very Low|
|45 - 54||40 - 44||W||Low|
|35 - 44||-||X||Moderate|
|25 - 34||-||Y||High|
|< 25||-||Z||Very High|
DRY FLOOR FRICTION SLIP RESISTANCE TEST
The dry floor friction slip resistance slip test (AS/NZS 4586 Appendix B) uses a battery operated machine commonly referred to as the "dry FFT" a or "Tortus". The dry FFT measures the force opposing the motion of a 9 mm diameter test slider of Four S rubber as it moves across the surfaces at a constant speed of 1m/min. The average of two measurements of 800 mm is taken and the results are expressed as Coefficient of Friction (CoF). The coefficient of friction is a ratio of the horizontal and vertical forces and the slip resistance test results are classified as follows:
Floor friction tester mean value
AS/NZS 4586 Classification
AS/NZS 4663 Notional* contribution of the floor surface to the risk of slipping when dry
Moderate to very low
High to very high
WET BAREFOOT RAMP SLIP RESISTANCE TEST
The wet barefoot ramp slip resistance test (AS/NZS 4586 Appendix C) uses a panel of the test surface which is constructed on a ramp large enough for a person to walk on. Two test people attach themselves to a safety harness, after bathing their feet for 10 minutes to soften the skin, walk bare foot on the test surface while the angle of the platform is increased until the limit of safe walking is achieved. This is generally when the test walker slips or feels that they will slip on the board. The angle that the limit of safe walking is obtained is known as the mean angle of inclination. Three calibration boards A, B & C are also tested, with the classification of the test panel based on the mean angle of inclination obtained in comparison with the calibration boards.
Mean Angle of
OIL WET RAMP SLIP RESISTANCE TEST
The oil wet ramp slip resistance test (AS/NZS 4586 Appendix D) differs to the wet barefoot ramp test, in that instead of water, an amount of motor oil is placed on the test surface and the two test persons wear safety boots rather than barefoot. Three calibration boards are walked on in a similar manner to the wet barefoot ramp test. These calibration boards are then used to provide a correction factor to standardise the walkers. Studies have shown that the correction factor reduces variance to 2.2 degrees (95% confidence interval). The slip resistance test classification is determined by the mean angle of inclination achieved and a correction factor based on the angle of inclination of the 3 calibration panels.
Mean Angle of
≥ 6 < 10
≥ 10 < 19
≥ 19 < 27
≥ 27 < 35
The information on this webpage was sourced from http://safeenvironments.com.au/slip-resistance-test-methods/. Safe Environments (www.SafeEnvironments.com.au) have provided Academy Tiles permission to reprint. For further information on slip resistance testing and consulting, please phone Safe Environments on 02 9624 2600; or email email@example.com.
The information on this webpage is based on Australian Standards, AS/NZS 4586 and AS/NZS 4663. These Australian Standards can be purchased online at www.saiglobal.com. Safe Environments Pty Ltd has acknowledged that permission to reprint has been given by SAI Global Ltd under Licence.